What are the two wavelengths used for Insertion Loss testing?
In standard Singlemode cable assembly, the two wavelengths used for Insertion Loss testing are 1310nm and 1550nm. All Singlemode fibers work very similarly in either wavelength—that is, you don’t need to buy fiber based on wavelength, one fiber fits all. So, IF your cable assembly is built properly, with good materials and good techniques, the Insertion Loss values for any given connector should be very similar when tested at either 1310 or 1550.
READ THE FULL BLOG ARTICLE HERE: Insertion Loss Troubleshooting Tip: Singlemode 1310 vs. 1550
ADDITIONAL CONTENT & RESOURCES:
- Category Resource: FOC Test Page
- Find More Information in this Blog: Testing Tips: enhance your process, your results, and your fiber optic cable assemblies
Answered by AskFOC Technical Team September 3, 2019
Have a technical question for Fiber Optic Center?
Please email your question to AskFOC@focenter.com and we will respond ASAP.
In addition to replying, we will post your question and our answer here on focenter.com. Everyone remains anonymous when we post so feel comfortable asking. You can also search your question and others in our content search.
- Considerations regarding freezing epoxy – THAWING - November 13, 2019
- Crystallization in Epoxies: How to Identify and Reverse it - October 23, 2019
- Measuring Insertion Loss and Return Loss Recommendations to Achieve the Best and Most Accurate Test Measurement - October 23, 2019
- Cross-Sectioning Fiber Optic Connectors: An Effective Diagnostic Method to Identify Defects and Resolve Process Issues - October 23, 2019
- The “Ideal” Fiber Height for a Fiber Optic Connector – and How to Achieve It - October 23, 2019
- Considerations regarding freezing epoxy – recommended length of time - October 23, 2019
- Techniques when using stripping tools - October 18, 2019
- Considerations regarding freezing epoxy – implementing process controls - September 18, 2019
- Aramid yarns and Good stripping techniques - September 11, 2019
- When the “back-side” connector end-face is contaminated, it is difficult to have access to remove the connector for cleaning - September 8, 2019