Where does the contamination come from in the normal polishing process?
Contamination is primarily a natural result of the normal polishing process. Ferrules are polished on abrasive films or flock-pile pads impregnated with abrasives. During polishing, some of these abrasive particles, along with the ceramic or plastic material being removed from the ferrule itself, come free and mix with the water used as lubricant within the process. This results in an abrasive slurry that contaminates the fixture and the ferrules themselves. Common polishing practices require the operator to clean the bottom of the fixture between polishing cycles, to remove abrasive particles that may create fiber end-face scratches in subsequent polishing steps. However, this slurry can (and will) migrate into crevices and the bores of the fixture—places difficult or unlikely to be fully cleaned or flushed out during normal the normal polishing process. Allowed to dry, it will harden and cake.
READ THE FULL BLOG ARTICLE HERE: Fiber Optic Polishing Fixture Maintenance: Fiber Optic Polishing Fixture Contamination and Ultrasonic Cleaning
Answered by AskFOC Technical Team August 16, 2018
Have a technical question for Fiber Optic Center?
Please email your question to AskFOC@focenter.com and we will respond ASAP.
In addition to replying, we will post your question and our answer here on the AskFOC page. Everyone remains anonymous when we post so feel comfortable asking your questions. You can also search to see if your question has already been answered in the FAQs section below.
- FOC Tips: Use the proper strip template when stripping for connectorization - July 17, 2019
- Fiber Optic Center AskFOC: which epoxy is the best for my application? - July 15, 2019
- Maple Syrup, Hidden Trails and Eerie Sights on the Third-Tallest Mountain in Vermont - July 14, 2019
- FOC Tips: the four major components of the fiber optic patch cord - July 12, 2019
- Fiber Optic Center AskFOC: when stripping multi-layer cables for connectorization, does each layer need to be stripped individually? - July 10, 2019
- FOC Tips: IL @ 1550 higher than 1310 - July 9, 2019
- Fiber Optic Center AskFOC: Should we check for crystallization? Do we need to adjust our inspection process? - July 8, 2019
- FOC Tips: diameter of the ferrule’s hole bore is larger than the outside diameter of the fiber - July 5, 2019
- Fiber Optic Center AskFOC: how do we establish good cleaning techniques throughout the polishing process? - July 3, 2019
- FOC Tips: Connectors are aligned and contacted optimally - July 2, 2019