From https://focenter.com/cross-sectioning-service/ Fiber Optic Connector Cross Sectioning and Analysis observations from “Connector C” notes
Observations: There is a substantial gap between the end of the stripped buffer area to the point of breakage (just inside the internal chamfer of the ceramic ferrule). Also evident is the shrinkage of epoxy which is quite apparent against the chamfer. It appears that the epoxy inside the ferrule cured first and acted as an anchor point during the thermal curing process.
As the thicker regions of epoxy were curing either accidental backward movement of the fiber pulled and fractured the fiber, creating the visible gap, or the rapid shrinkage of the epoxy pulled everything and “separated” from the anchored fiber inside the ferrule.
Additional resources from the FOC team include:
- Category Resource
- Glossary, Acronyms, Military Specifications for Connectors: http://bit.ly/2a2EFn8
- Q&A Resource: email technical questions to AskFOC@focenter.com
Have questions about this FOC Tip?
Contact FOC with questions at: (800) 473-4237 / 508-992-6464 or email: FiberOpticCenter@focenter.com and we will respond ASAP.
- Considerations regarding freezing epoxy – THAWING - November 13, 2019
- Crystallization in Epoxies: How to Identify and Reverse it - October 23, 2019
- Measuring Insertion Loss and Return Loss Recommendations to Achieve the Best and Most Accurate Test Measurement - October 23, 2019
- Cross-Sectioning Fiber Optic Connectors: An Effective Diagnostic Method to Identify Defects and Resolve Process Issues - October 23, 2019
- The “Ideal” Fiber Height for a Fiber Optic Connector – and How to Achieve It - October 23, 2019
- Considerations regarding freezing epoxy – recommended length of time - October 23, 2019
- Techniques when using stripping tools - October 18, 2019
- Considerations regarding freezing epoxy – implementing process controls - September 18, 2019
- Aramid yarns and Good stripping techniques - September 11, 2019
- When the “back-side” connector end-face is contaminated, it is difficult to have access to remove the connector for cleaning - September 8, 2019