For Bi-Directional testing, can I run the test in both direction and not have to swap the cable and run the test again?
It’s fair to say most assembly houses test the optical loss of a single mated pair using a launch cable connected to the DUT with the far end of the DUT plugged directly into the power meter via the adapter cap. To complete the test, the DUT is flipped and the second connector is then tested. Here is where people envision the time savings. Won’t it be great to run the test in both direction and not have to swap the cable and run the test again? Well yes, but here’s the rub: you have one side attached without a launch. Bi-Directional test requires the use of a launch AND a receive cable, so when it light it shot in one direction there is a launch and when it’s shot in the opposite direction there is also a launch.
So if you aren’t already using the launch and receive cable method of testing you’ll have to change your test method to fully utilize the Bi-Directional feature. The launch and receive jumper method for optical loss give the loss of the entire jumper in one direction.
READ THE FULL BLOG ARTICLE HERE: Is Bi-Directional testing the silver bullet to cut all your optical testing time in half?
ADDITIONAL CONTENT & RESOURCES:
- Category Resource: FOC Test Page
- Find More Information in this Blog: Testing Tips: enhance your process, your results, and your fiber optic cable assemblies
Answered by AskFOC Technical Team September 28, 2018
Have a technical question for Fiber Optic Center?
Please email your question to AskFOC@focenter.com and we will respond ASAP.
In addition to replying, we will post your question and our answer here on focenter.com. Everyone remains anonymous when we post so feel comfortable asking. You can also search your question and others in our content search.
- Considerations regarding freezing epoxy – implementing process controls - September 18, 2019
- Aramid yarns and Good stripping techniques - September 11, 2019
- When the “back-side” connector end-face is contaminated, it is difficult to have access to remove the connector for cleaning - September 8, 2019
- Fiber Optic Cable Splicing Explained - September 7, 2019
- Ways in which the end-face can become contaminated along the way - September 6, 2019
- Removal of coatings from short fiber in certain sections - September 5, 2019
- End-face quality and cleanliness - September 4, 2019
- Re-cleaning a connector after installation - September 4, 2019
- The two wavelengths used for Insertion Loss testing - September 3, 2019
- Ensuring connector end-faces meet industry standards on end-face cleanliness (ex. IEC 61300-3-35) - September 2, 2019