We have had no changes in our fiber optic cable assembly production process, but the measured Insertion Loss (IL) and Return Loss (RL) values of our product aren’t as good as they were. Could there be something wrong with the measurement equipment?
Obviously something has changed. If it’s not the measurement equipment, then it must be either the product or the measurement setup. Over time the production process can become a familiar, perhaps boring, routine. Operators may become less meticulous in cleaning, use polishing film longer than appropriate, or even take shortcuts. Eventually this takes a toll and yield suffers.
Let’s suppose you’ve eliminated this possibility with a clean sweep of procedures and a reset to rigorous production guidelines. Yet the problem persists. A “gold-standard” product is useful at this stage – a “known-good” example of the best of your previous production, carefully kept, with very good IL and RL values recorded. Using your current rigorous procedures, re-measure this gold-standard product. It should still look good. If it doesn’t, read on …
I have written that IL and RL tests on a product cannot be made in isolation. (Read more on this topic here.) Remember, we are measuring connector loss, and the device under test (DUT) needs to be connected to a test cable (aka jumper cable, reference cable, or test lead). It’s important to note that the quality and condition of test cables have a direct bearing on the measured product’s IL and RL measurements.
READ THE FULL BLOG ARTICLE HERE: Test cables don’t last forever: How to assess quality, condition, usage limit, and replacement schedule
ADDITIONAL CONTENT & RESOURCES:
- Category Resource: FOC Test Page
- Find More Information in this Blog: Testing Tips: enhance your process, your results, and your fiber optic cable assemblies
Answered by AskFOC Technical Team April 17, 2019
Have a technical question for Fiber Optic Center?
Please email your question to AskFOC@focenter.com and we will respond ASAP.
In addition to replying, we will post your question and our answer here on focenter.com. Everyone remains anonymous when we post so feel comfortable asking. You can also search your question and others in our content search.
- Techniques when using stripping tools - October 18, 2019
- Considerations regarding freezing epoxy – implementing process controls - September 18, 2019
- Aramid yarns and Good stripping techniques - September 11, 2019
- When the “back-side” connector end-face is contaminated, it is difficult to have access to remove the connector for cleaning - September 8, 2019
- Fiber Optic Cable Splicing Explained - September 7, 2019
- Ways in which the end-face can become contaminated along the way - September 6, 2019
- Removal of coatings from short fiber in certain sections - September 5, 2019
- End-face quality and cleanliness - September 4, 2019
- Re-cleaning a connector after installation - September 4, 2019
- The two wavelengths used for Insertion Loss testing - September 3, 2019