One way to characterize oven ports is by using a thermocouple probe inserted and potted with epoxy into a ferrule – just like when terminating a fiber, except you’ll be inserting the thermocouple tip instead of a fiber.
Cure it, as you would a regular termination, and now you have a temperature probe which mimics your product exactly.
For example, if you pot the thermocouple into an SC connector, now you have a thermocouple probe with an SC connector on one end, and your measurements will indicate the temperature INSIDE the connectors, which is really the temperature we are most concerned with.
Follow these steps:
- Use a 0.9-millimeter (900-micron) diameter thermocouple – or smaller, to ensure the thermocouple tip can fit inside the rear of a ferrule or connector you will use.
- Inject epoxy into the ferrule (or connector), and pot the thermocouple probe in the ferrule, so that the thermocouple tip is bottomed out against the base of the ferrule. (Do this as if you’re making a normal connector with optical fiber. Of course, there will be no fiber or anything within the ferrule bore, just the tip of the thermocouple positioned in the middle of the ferrule assembly.)
- Cure as normal. This will be your temperature probe: a ferrule (or connector) with thermocouple tip positioned in the center of the ferrule, right where the fiber strip point will be in your product. This is the area you want to ensure reaches your curing temperature set point.
- Place a thermocouple-connector probe into each port of your curing oven. Note how much time is needed for the center of the ferrule to reach the desired curing temperature, what the maximum temperature reached is, and how stable the temperature is throughout the curing time.
- Repeat the above steps to measure the temperature in every port. Since temperatures can vary port-to-port, this process will tell you whether the temperature is consistent across all ports in your curing oven.
- Characterize the oven ports annually, so you can track whether the temperature in all ports has remained stable over the past year. Adjust your curing schedule, or replace the curing oven as necessary.
Additional resources from the FOC team include:
- Glossary, Acronyms, Military Specifications for Connectors: http://bit.ly/2a2EFn8
- Q&A Resource: email technical questions to AskFOC@focenter.com
- Contact FOC with questions at: (800) 473-4237 / 508-992-6464 or email: firstname.lastname@example.org
- Fiber Optic Center AskFOC: where does the contamination come from in the normal polishing process? - August 16, 2018
- FOC Tips: spring loaded connectors after primary polishing - August 15, 2018
- Fiber Optic Center AskFOC: how are the environments in which copper and optical cable are produced different? - August 14, 2018
- FOC Tips: inner-housing - August 13, 2018
- FOC Tips: the most common fiber optic epoxy packaging is premeasured two part packs. - August 11, 2018
- FOC Tips: 3D printed objects start with a good print job - August 10, 2018
- Fiber Optic Center AskFOC: what are the advantages and disadvantages of the different methods of cross sectioning fiber optic connectors? - August 9, 2018
- Fiber Optic Center AskFOC: what lapping films are available with PSA (Pressure Sensitive Adhesive) back? - August 7, 2018
- FOC Tips: To help ensure accurate RL measurements, use OTDR. - August 3, 2018
- Fiber Optic Center AskFOC: When does less cladding get faster transmission rates or does it? - August 2, 2018